MET ONE 6005 Remote, Down Exhaust, Ethernet Interface, No Flow Meter
|Performance||ISO 21501-4 unit-to-unit accuracy & reproducibility|
|Industry Standards||ISO 21501-4 , ISO 14644-1:2015, FCC, CE|
|Light Source||Long Life Laser™ Diode|
|Particle Size Analysis Range||0.5 - 10 μm|
|Flow Rate||2.83 L per minute|
|Coincidence Loss||10% at 140,000,000 particles /m³|
|Sample Volume||10 µm|
|Throughput||2.83 L per minute|
|Interfaces & Protocols||Ethernet with Modbus TCP protocol|
|Data Storage||1,000 samples (oldest records overwritten when buffer is full)|
|Zero Count||ISO 21501-4 & JIS B 9921 (1 count or less in 5 min; 95% UCL)|
|Count Alarms, Display & Cycles||User-defined alarm counts displayed via LED on box or remote LED|
|Environmental Conditions||Indoor Environment|
|Temperature Range||Operating temperature range: 10°C to 32°C, Storage temperature range: -40 to +70°C|
|Humidity Restrictions||5% - 85% (non-condensing)|
|Power Requirements||4.3 W, 9 to 28 VDC|
|Software||Integrates with various software platforms for facility monitoring|
|Operating Modes||Remote software-controlled sampling|
|Weight||0.8 kg(1.8 lb)|
|Width||13.6 cm(5.4 in)|
|Depth||8.9 cm(3.5 in)|
|Height||12.1 cm(4.8 in)|
NOT ALL PRODUCTS ARE AVAILABLE IN ALL COUNTRIES
PRODUCT AVAILABILITY AND REGULATORY STATUS DEPENDS ON COUNTRY REGISTRATION PER APPLICABLE REGULATIONS
The listed regulatory status for products correspond to one of the below:
IVD: In Vitro Diagnostic Products. These products are labeled "For In Vitro Diagnostic Use."
ASR: Analyte Specific Reagents. These reagents are labeled "Analyte Specific Reagents. Analytical and performance characteristics are not established."
CE: Products intended for in vitro diagnostic use and conforming to European Directive (98/79/EC). (Note: Devices may be CE marked to other directives than (98/79/EC)
RUO: Research Use Only. These products are labeled "For Research Use Only. Not for use in diagnostic procedures."
LUO: Laboratory Use Only. These products are labeled "For Laboratory Use Only."
No Regulatory Status: Non-Medical Device or non-regulated articles. Not for use in diagnostic or therapeutic procedures.