|Detection||Photodiodes (forward & side scatter), Photomultiplier tubes (fluorescence)|
|Resolution||0.5 μm diameter particles. HPCV is <2% to <4% depending on the laser.|
|Performance||<600 MESF for FITC, <300 MESF for PE, <600 MESF for APC|
|Optics/Laser Power||488 nm argon ion laser (20 mW)|
|Flow Cell||BioSense 150 μm x 450 rectangular channel with an integral lens|
|Signal Processing||Digital System with 4 decade data display|
|Power Requirements||100-240 VAC, 50/60 Hz|
|Temperature Range||16°C - 32°C|
|Humidity Restrictions||35%-85% (non-condensing)|
|Installation Category||II (per IEC 1010-1 standard)|
|Weight||84.8 kg / 187 lb|
|Width||97.8 cm / 38.5 in|
|Depth||88.9 cm / 35 in|
|Height||61 cm / 24 in|
NOT ALL PRODUCTS ARE AVAILABLE IN ALL COUNTRIES
PRODUCT AVAILABILITY AND REGULATORY STATUS DEPENDS ON COUNTRY REGISTRATION PER APPLICABLE REGULATIONS
The listed regulatory status for products correspond to one of the below:
IVD: In Vitro Diagnostic Products. These products are labeled "For In Vitro Diagnostic Use."
ASR: Analyte Specific Reagents. These reagents are labeled "Analyte Specific Reagents. Analytical and performance characteristics are not established."
CE: Products intended for in vitro diagnostic use and conforming to European Directive (98/79/EC). (Note: Devices may be CE marked to other directives than (98/79/EC)
RUO: Research Use Only. These products are labeled "For Research Use Only. Not for use in diagnostic procedures."
LUO: Laboratory Use Only. These products are labeled "For Laboratory Use Only."
No Regulatory Status: Non-Medical Device or non-regulated articles. Not for use in diagnostic or therapeutic procedures.